The ideal test environment is a quiet room. High ambient background noise adversely affects OAE measurements.
The baby should be sleeping or in quiet relaxed state. Sucking, blinking, crying or movement may affect testing.
Place the probe in the ear – the light will turn green when a good seal is obtained.
Click on START in the software, press the spacebar or press the shoulder box button.
PASS – when the criteria for a pass are reached, PASS is displayed in green above the measurement. REFER – when the criteria for a pass are not reached within the measurement time, REFER is displayed in amber above the measurement.
INCOMPLETE – if the test is stopped before a PASS or REFER is generated by the system, INCOMPLETE is displayed above the measurement indicating that the full test was not completed.