Hearing Aid Fitting
Hearing Aid Fitting
The Variety of Decibel
Tips for the use of Sanibel disposable Tab and Snap electrodes
2nd Generation ASSR
How to Prepare a Baby for Automated ABR
Daily System Check
IA OAE Suite TEOAE
IA OAE Suite DPOAE
Electrode Tips & Tricks
Neuro Latency Testing
Rate Study Protocol
Middle Latency Response (MLR)
Fmp and Residual Noise
How to Improve Evoked Potential Recordings
Electrical Auditory Brainstem Response (eABR)
Basic ABR Testing
ALR / Cortical Evoked Response Audiometry
ASSR Hearing Test
How to mark ABR waveforms
How to make changes to a saved session
How to add a new patient in OtoAccess® Database
How to print a report
How to customize your ABR protocol
How to use the latency tab
How to use the recording tab
How to use the top panel
How to use the edit tab
How to navigate the test screen
Mounting the Eartip Adaptor
15 February 2022
Remove the nipple from the insert phone tubing.
Mount the eartip adaptor ensuring it is inserted into the tubing completely. Place an eartip on the adaptor before use.
Ensure that the eartip adaptor is not blocked with vernix or cerumen after use. For cleaning, follow the instructions below.
Cleaning the eartip adaptor
Unplug the eartip adaptor from the insert phone tubing and use the cleaning floss that comes with the adaptor kit (8003566).
Thread the stiff end of the cleaning floss into the eartip adaptor tube from the back. Now pull the cleaning floss completely through tube. Repeat if necessary.
If the adaptor is severely contaminated with cerumen or vernix, replace the adaptor with a new one (part no. 8500580) to ensure correct usage.
Popular Academy Training
Getting started: ASSR
This course provides an overview of the basics of performing the auditory steady state response (ASSR) test.
What is nHL-to-eHL Correction?
nHL-to-eHL Correction for ABR Stimuli
Course: An Introduction to the CE-Chirp®
This course provides an introduction to the CE-Chirp® stimulus family, which is found in the Interacoustics Eclipse and Titan platforms.
Find a distributor
Hearing aid fitting systems
Balance testing equipment