The quick switch... switch now and clean later...
Step 1: Unscrew the probe cap.
Step 2: Remove the probe tip. Make sure there is a gasket in the clean probe tip.
Step 3: Replace with a clean/replacement tip.
Step 4: Assemble the probe again (step 2+1).
Clean the probe tip regularly: To avoid earwax clogging the tubes of the probe tip and affecting measurements, it is recommended that you clean the probe tip regularly.
Complementary probe tip enclosed: Please find a complementary probe tip enclosed with your system purchase.
Easily swap the probe tip: Easily swap the probe tip during the day, and clean it when time permits.
Please follow the steps 1 to 4 when you remove the probe tip and replace it with a new, clean one.